Can I use a test strip from another Accu-Chek system in the Accu-Chek Performa blood glucose meter?
No. Accu-Chek Performa blood glucose meters are designed exclusively for use with Accu-Chek Performa test strips.
Can I use an expired test strip?
No, check the use by date on the test strip container. Do not use test strips past the use by date.
When and why is it helpful to test with a control solution?
Performing a control test lets the user know the meter and test strips are working properly. Perform a control test when: You open a new test strip box You left the test strip container open You think the test strips are damaged You want to check the meter and test strips The test strips were stored in extreme temperatures, humidity, or both You dropped the meter Your test result does not match how you feel You want to check if you are performing the test correctly
How do I run a control test?
The Accu-Chek Performa blood glucose meter automatically recognizes Accu-Chek Performa Control Solutions, so running a control test is easier than ever. First, wipe the tip of the control bottle. Then put the meter on a flat surface, insert the strip, and squeeze a drop of the control solution onto the front edge of the test strip. A result appears with a flashing “L”. Press one of the arrow keys once to indicate a Level 1 control test or twice to indicate Level 2. Press the ON/OFF button to set the level in the meter. The control result and “OK” alternate in the display if the result is in range.
Where can I find out what an error code or screen message is telling me?
Visit your meter' Screen Messages page for a list of messages and definitions or download the User Manual for your meter.
How do I set or change the time and date on my meter?
Watch this short video to see how easy it is to set the date and time. Your meter comes with the batteries already installed and a preset time and date. You may need to change the time to match your time zone or, if you live in a state that observes daylight saving time, to reflect the time change. To change these settings: Press and release the On/Off button to turn the meter on. The flashing test strip symbol appears. NOTE: If this is the first time your meter has been turned on, it will automatically enter the set-up mode, and "Set-up" and the hour will flash on the display. Skip to Step 3 to change the time or, if the date and time are correct, press and hold the On/Off button to exit the set-up mode. To enter the set-up mode, press and hold the On/Off button for about four seconds. “Set-up” and the hour flash on the display. Press and release the arrows to decrease or increase the hour. Press and hold the arrows to scroll faster. Press and release the On/Off button to set the hour. The minutes flash. Repeat steps 3 and 4 to set the minutes, AM/PM, month, day and year. The flashing field is the one you are changing. Once you set the correct time and date, press and hold the On/Off button until the flashing test strip symbol appears.
How do I change the battery in my Accu-Chek Performa blood glucose meter?
Make sure you have a new 3-volt, lithium battery, number CR2032, to insert. Open the battery door on the back of the meter, remove the old battery for at least 20 seconds, and press any button on the meter. Insert the new battery with the “+” sign facing out. Snap the battery door back into place. Press the on/off button, and the meter will beep. Setup and hour are flashing on the display. If the date and time are correct, press and hold the on/off button for about 4 seconds. Release the button, and you’re done.
What Accu-Chek products can be used with Accu-Chek 360° software?
Accu-Chek Active meter Accu-Chek Performa meter Accu-Chek Aviva meter Accu-Chek Nano SmartView system Accu-Chek Performa Nano meter Accu-Chek Compact meter Accu-Chek Go meter Accu-Chek Integra meter Accu-Chek Compact Plus meter Accu-Chek Mobile meter Accu-Chek Aviva/Performa Expert meter Accu-Chek Insight system Accu-Chek Insight Aviva/Performa diabetes manager Accu-Chek Aviva/Performa Connect meter Accu-Chek Voicemate Plus meter Accu-Chek Spirit insulin pump Accu-Chek Combo system Accu-Chek Spirit Combo insulin pump Accu-Chek D-TRON plus insulin pump Accu-Chek Connect Online system
What do the error signals mean?
The Accu-Chek Smart Pix device reader reports potential errors during data transfer and use by causing the center display section 8 to flash. Here's what you can do: In the browser navigation, click the Read Device button. The center display section should flash slowly now, indicating that it is actively searching for devices and that it is ready for data transfer. If the error signal continues, you can unplug the Accu-Chek Smart Pix device reader and plug it back in again. Disregard any Windows error messages about removing the data medium. Try repeating the transfer again. If this error signal occurs again, check the following: Has the meter been properly prepared for data transfer? Is there anything obstructing the signal between the device IR port and the Accu-Chek Smart Pix device reader? Is the device too far away? Is there any interference from bright lights or sunlight? Often, in the case of error messages associated with data transfer, the issue is interference with data transfer, or a problem with the meter or insulin pump.
What software can I use to download data from my Accu-Chek Combo system?
The system is compatible with the Accu-Chek 360° diabetes management system, Accu-Chek Connect and the Accu-Chek Smart Pix device reader.
Do I need to code the Accu-Chek Aviva Combo meter?
The Accu-Chek Aviva/Performa Combo meter offers simple 1-time coding. You'll see that the red-and-white test strip package includes a black code key. Your meter should already have a black code key inserted. Just leave it in place. Your meter is already coded.
What are the main differences between the old and the new Accu-Chek Active test strips?
There are 2 main differences: The Accu-Chek Active test strip uses a new chemistry, eliminating clinically relevant interferences with maltose. The color of both the test area on the test strip and code chip is now green.